|Components of MRI||Effect on PET||Consequences||Resolution|
|Static magnetic field (B0)||Influences the ray of electron flow in read out.||Lower gain factor.||Substitute the Photomultiplier with Solid State Photo Detector (SSPD) such as SiPM or APD.|
|Gradient magnetic field (B1)||Heats up the equipment leading to vibrations.||Induction of eddy currents and mechanical vibrations.||
Restructuring of electronic components.|
Fixing temperature control device.
|RF Interference||Introduces noise signal.||
Due to magnetic induction, introduces eddy current.|
Produces heating effect.
|Introducing RF shield around the coil.|